Supplementary data: High sensitivity micro-volume UV absorption spectrometry for routine analysis of small volume biological samples
Supplementary Table S1. Comparison of sample volume, path length, and detection limit among commercial nano- or micro-volume absorption spectrometers and apparatus. Data in the table were obtained from manufacturers’ webpages (refer to the references indicated in the table). The results of this work are listed at the bottom line for comparison.
Supplementary Table S2. Information on the internal volumes of micro-tubes with various combinations of internals dimeters and lengths.
Supplementary Figure S1. Schematic of the fluidic control body to be assembled with the spacer defining the internal fluidic channel. The fluidic control body also provides receptacles for an absorption cell as well as an optical fiber fetch cable to optically couple them together.
Supplementary Figure S2. The detailed presentation of the interfacing area of the fluidic control body.
Supplementary Figure S3. Schematic of the light passage for absorption measurement (not to scale): (a) light source, (b) optical fiber, (c) fiber receptacle, (d) window, (e) absorption cell, (f) detection well, (g) spectrophotometer module or photodiode.
Supplementary Figure S4. Graphical presentation of the prototype instrument equipped with the proposed absorption cell as well as the disk type sample tray.